Pump+setup+instructions

This May not Make a lot sense at this point unless you see the pump. Anyways it could be useful during the rotation.

- [ ] Set up reservoir - [ ] Do not place the reservoir box on the floor - [ ] Remove the following caps before mounting the reservoir - [ ] Water line caps - [ ] Gas line cap (only if the circuit is going to be used immediately) - [ ] White cap (Manufacturer sterilization) - [ ] Temp probe caps - [ ] Stick the Level sensor - level 250 - [ ] Attach temp probes (no color coding) and level detector - [ ] Attach heater cooler. "In" at the top, "Out" at the bottom - [ ] Open up the pump pack - [ ] Save the drapes. - [ ] Take the tubings (arterial, yellow - sucker, green - vent) lines and keep them aside. - [ ] Attach the CO2 line (90 degrees) to the arterial line filter - [ ] Position the arterial line filter in such a way that its easy to connect the outlet of the oxygenator to the filter - [ ] ALWAYS when you are dealing with a sterile connection, Make the stationary object vulnerable. - [ ] Clamp the line that is hanging on the floor so that it is not contaminated - [ ] Connect the arterial line filter to the outlet of the oxygenator. - [ ] Clamp the outlet of the filter. Position the clamp in such a way that no one can knock it out. - [ ] Attach the pressure separator. - [ ] Attach the quick prime line (white middle cap in the center of the reservoir) - [ ] Clamp one of the arms of the quick prime line - [ ] Insert the arterial line tubing in the roller head. - [ ] First attach one end to the venous reservoir - [ ] Attach the other end to the oxygenator - [ ] Then mount the tube in the raceway - [ ] Follow reverse roller head roll technique for proper seating - [ ] Insert the sucker line (Yellow - 1) - [ ] Follow reverse roller head roll technique - [ ] Do not Make the "reservoir end" connection tight - [ ] Insert the vent line (Green - 2) - [ ] Follow reverse roller head roll technique - [ ] Do not Make the "reservoir end" connection tight - [ ] Take the stuff for the nurses and put it on the chair - [ ] Cardioplegia line setup - [ ] Attach the pop off valve - [ ] Attach the pressure separator (Make sure the direction of the pressure separator is correct) - [ ] Slide the cardioplegia setup in the console - [ ] Attach the other end of the pop off valve to the reservoir - [ ] Leave a 6" slack and put the tubes in the raceway - [ ] Its easier to put both the raceways simultaneously. - [ ] Connect heater cooler to cardioplegia setup.